Sterilizing a vial
- 08-17-2014, 06:15 PM
- 08-17-2014, 07:37 PM
- 08-17-2014, 08:10 PM
09-08-2014, 07:49 AM
It depends on the compound too. Tren will break down at a lower temp. So you can't go over 175 degrees I believe with Tren. All others should be fine at 240-250 degrees. Make sure you put a large gauge needle w/o syringe in the top of vial to vent.
09-08-2014, 02:38 PM
I would think the higher heat could damage the rubber no? get or borrow an autoclave > http://en.wikipedia.org/wiki/Autoclave IMO
LG Sciences Board Rep
These statements have not been evaluated by the FDA, do not constitute medical advice, and are not official or authorized comments by LG Sciences, LLC.
09-08-2014, 08:25 PM
He's talking about the vials. It doesn't matter what the temp is for the glass vials. The rubber is good at 220, but I hesitate to go any higher. I think you are thinking he is mixing the compounds. He's just sterilizing the glass and stopper, from what I understand.
09-08-2014, 08:27 PM
I did a half hour at 220 awhile back and it turned out fine. No issues with melting rubber or anything. Did have some spillage thru the needle which i assume is from the pressure increase tho. Prob lost a ml or so..
Similar Forum Threads
- By Lockxxheed in forum Cycle InfoReplies: 0Last Post: 08-30-2011, 09:51 PM
- By orangutan in forum AnabolicsReplies: 4Last Post: 02-23-2009, 05:06 AM
- By sicnarf in forum AnabolicsReplies: 12Last Post: 07-07-2006, 10:36 PM
- By Viator in forum AnabolicsReplies: 4Last Post: 02-14-2005, 07:02 PM